Hioki 333603 전력 품질 분석기 (2ch, GPIB, D / A 출력)
 연락처
SPECIFICATIONS
 Measurement line type:
+ Singlephase 2wire, singlephase 3wire
+ Threephase 3wire
 Measurement items:
+ Voltage, Current, Active power, Apparent power, Reactive power
+ Power factor, Phase angle, Frequency, Efficiency
+ Current integration, Active power integration, Integrated time
+ Voltage waveform peak value, Current waveform peak value
+ Voltage crest factor, Current crest factor, Time average current
+ Time average active power, Voltage ripple factor
+ Current ripple factor
 Measurement range:
+ Voltage range: AC/DC 15 V to 1000 V, 7 ranges
+ Current range: AC/DC 200 mA to 50 A, 8 ranges
+ Power range: 3.0000 W to 150.00 kW
 Basic accuracy(Active power):
+ ±0.1% rdg. ±0.1% f.s. (DC)
+ ±0.1% rdg. ±0.05% f.s. (45 Hz to 66 Hz, at Input < 50% f.s.)
+ ±0.15% rdg. (45 Hz to 66 Hz, at 50% f.s. ≤ Input)
 D/A output:
+ 16 channels, Level output DC ±2 V, Waveform output 1 V f.s
+ Level output, instantaneous waveform output(U,I,P)
+ Level output (apparent power, reactive power, power factor, or other)
+ Highspeed active power level output
 Other functions:
+ AC+DC, AC+DC Umn, AC, DC, FND, Autorange, Average
+ VT or CT ratio settings, Synchronized control
+ MAX/ MIN, or other functions
 External interfaces:
+ RS232C, GPIB
+ LAN standard
 Power supply: 100 to 240 V AC, 50/60 Hz, 40 VA max
 Product Net Weight: 5.6 kg
 Product Size: 305*132*256mm
 Supplied accessories:
+ Instruction manual ×1
+ Measurement guide ×1
+ Power cord ×1
 The warranty period: 12 month
Detail
Measurement lines  Singlephase 2wires, singlephase 3wires, 3phase 3wires, (voltage / current measurement range set for each wiring mode) 
Measurement items  Voltage, Current, Active power, Apparent power, Reactive power, Power factor, Phase angle, Frequency, Efficiency, Current integration, Active power integration, Integrated time, Voltage waveform peak value, Current waveform peak value, Voltage crest factor, Current crest factor, Time average current, Time average active power, Voltage ripple factor, Current ripple factor 
Harmonic parameters  Synchronization frequency range: 10 Hz to 640 Hz, Analysis order up to 50th Harmonic voltage RMS value, Harmonic current RMS value, Harmonic active power, Total harmonic voltage distortion, Total harmonic current distortion, Voltage fundamental waveform, Current fundamental waveform, Active power fundamental waveform, Apparent power fundamental waveform, Reactive power fundamental waveform, Power factor fundamental waveform (displacement power factor), Voltage current phase difference fundamental waveform, Interchannel voltage fundamental wave phase difference, Interchannel current fundamental wave phase difference, Harmonic voltage content % , Harmonic current content % , Harmonic active power content % (The following parameters can be downloaded as data during PC communication but not displayed: Harmonic voltage phase angle, Harmonic current phase angle, Harmonic voltage current phase difference) 
Measurement range  Voltage range: AC/DC 15 V to 1000 V, 7 ranges Current range: AC/DC 200 mA to 50 A, 8 ranges Power range: 3.0000 W to 100.00 kW (Depends on combination of voltage and current range) 
Integration measurement (Integration time up to 10,000 hours) 
[Current] No.of displayed digits: 6 digits (from 0.00000 mAh, Polarityindependent integration and Sum value) [Active power] No.of displayed digits: 6 digits (from 0.00000 mWh, Polarityindependent integration and Sum value) 
Input resistance (50/60 Hz)  Voltage: 2 MΩ, Current: 1 mΩ or less (direct input) 
Basic accuracy (Active power)  ±0.1% rdg. ±0.1% f.s. (DC) ±0.1% rdg. ±0.05% f.s. (45 Hz to 66 Hz, at Input < 50% f.s.) ±0.15% rdg. (45 Hz to 66 Hz, at 50% f.s. ≤ Input) 
Display refresh rate  5 times per second to 20 second (depend on average times settings) 
Frequency characteristics  DC, 0.1 Hz to 100 kHz 
D/A output (02/03 model only)  16 channels (selectable from following items), Level output DC ±2 V, Waveform output 1 V f.s. Level output, instantaneous waveform output (voltage, current, active power) Level output (apparent power, reactive power, power factor, or other) Highspeed active power level output 
Functions  [Rectification method] AC+DC, AC+DC Umn, AC, DC, FND, Autorange, Average, VT or CT ratio settings, Synchronized control, MAX/ MIN, or other functions 
Interface  RS232C / LAN standard, (01/03 model also includes GPIB) 
Power supply  100 to 240 V AC, 50/60 Hz, 40 VA max. 
Dimensions and mass  305 mm (12.01 in) W × 132 mm (5.20 in) H × 256 mm (10.08 in) D, 5.2 kg (183.4 oz) 
Accessories  Instruction manual ×1, Measurement guide ×1, Power cord ×1 
OPTION






Current measurement [Type 2]  
Requires Sensor unit 955510 and Connection cord L9217 (1 sensor necessary for singlephase measurement, and 2 sensors required for 3phase measurements) 














PC communication  


Other options  
Synchronized control with the 9165 

DOWNLOAD CATALOG 
POWER METER PW3336/PW3337 product catalog.pdf (2.12 MB) (20130626 updated) 
 품질 보증
 공인 보증
 집으로 배달
 간단하게 거래하기